Read Before Using. Lire avant usage. Leer antes de usar. General Safety Rules. DO NOT stare into the laser light source.
|Published (Last):||17 February 2014|
|PDF File Size:||1.31 Mb|
|ePub File Size:||6.96 Mb|
|Price:||Free* [*Free Regsitration Required]|
Quick Links. Table of Contents. Instrucciones de funcionamiento y seguridad. DO NOT stare into the laser light source. Never aim light at another person or object other than the workpiece. Laser light can damage your eyes. Read all instructions. Page 7: Safe Operating Procedures unintentionally blind other persons. Repairs performed liquids, gases or dust. Page 11 Automatic switch-off after approx.. Use only alkali-manganese or rechargeable batteries. Always replace all batteries at the same time.
Only use Fewer measurements are batteries from one brand and possible when using 1. Measuring Procedure The measured value appears The rangefinder offers a variety after 0. Page Measuring Modes To select the reference point, push button Each Do not point the time after switching on, the rear laser beam at persons or edge of the rangefinder is preset animals and do not look into Page 15 Area Measurement Afterwards, measure the length, width and the height, one after For area measurements, push another, in the same manner as button 5 until the indicator for for a length measurement.
The area measurement laser beam remains switched on appears in the display. Move the laser back and forth To start the measurement, briefly push the measuring button Page 17 target surface is available as a After completing the second reflector. Correct results are measurement, the distance B is achieved only when the laser calculated automatically.
Page 18 at the bottom in the display, while the rangefinder can be moved the surface is indicated at the top. As an example, the user can move from a wall to the required distance, while the actual distance can be read Page 19 all measuring functions. Push the memory retrieve button 2 in order to display the value stored in the memory. Page 21 with varying temperatures or surfaces, especially over longer indirectly received reflections can distances.
These In combination with the laser effects are due to physical beam, the vial level 6 is not reasons and can therefore not suitable for leveling. Page Maintenance And Service Issue Remedy Measuring result not plausible Wrong reference point set Select reference point that corresponds to measurement Obstruction in path of laser beam Laser point must be completely on target surface The rangefinder section with a length of approx..
If you are not completely satisfied with the performance of your laser or measuring tool product, for any reason, you can return it to BOSCH within 30 days of the date of purchase for a full refund or replacement.
To obtain this Day Refund or Replacement, your return must be accompanied by the original receipt for purchase of the laser or measuring tool product. Print page 1 Print document 25 pages. Cancel Delete. Sign In OR. Don't have an account? Sign up! Restore password. Upload from disk. Upload from URL.
Bosch Laser Distance Measurer GLR225 PDF manual
Bosch GLR225 Operating/Safety Instructions Manual
Bosch GLR225 Manuals
BOSCH GLR225 Manual